Pitot tubes can be used to indicate fluid flow velocities by measuring the difference between the static and the dynamic pressures in fluids
A pitot tube can be used to measure fluid flow velocity by converting the kinetic energy in a fluid flow to potential energy.
The principle is based on the Bernoulli Equation where each term of the equation can be interpreted as pressure
p + 1/2 ρ v2 + γ h = constant along a streamline (1)
p = static pressure (relative to the moving fluid) (Pa)
ρ = density (kg/m3)
v = flow velocity (m/s)
γ = ρ g = specific weight (N/m3)
g = acceleration of gravity (m/s2)
h = elevation height (m)
Each term of the equation has the dimension force per unit area N/m2 (Pa) - or in imperial units lb/ft2 (psi).
The first term - p - is the static pressure. It is static relative to the moving fluid and can be measured through a flat opening in parallel to the flow.
The second term - 1/2 ρ v2 - is called the dynamic pressure.
The third term - γ h - is called the hydrostatic pressure. It represent the pressure due to change in elevation.
Since the Bernoulli Equation states that the energy along a streamline is constant, (1) can be modified to
p1 + 1/2 ρ v12 + γ h1
= p2 + 1/2 ρ v22 + γ h2
= constant along the streamline (2)
suffix1 is a point in the free flow upstream
suffix2 is the stagnation point where the velocity in the flow is zero
In a measuring point we regard the hydrostatic pressure as a constant where h1 = h2 - and this part can be eliminated. Since v2 is zero, (2) can be modified to:
p1 + 1/2 ρ v12 = p2 (3)
v1 = [2 (p2 - p1) / ρ] 1/2 (4)
p2 - p1 = dp (differential pressure)
With (4) it's possible to calculate the flow velocity in point 1 - the free flow upstream - if we know the differential pressure difference dp = p2 - p1 and the density of the fluid.
It is common to use head instead of pressure. (4) can be modified to
v1 = c [2 g (h2 - h1)]1/2 (5)
c = coefficient
g = acceleration of gravity (m/s2)
h2 - h1 = height difference (m)
The Pitot Tube
The pitot tube is a simple and convenient instrument to measure the difference between static, total and dynamic pressure (or head).
The head - h - (or pressure difference - dp) can be measured and calculated with u-tube manometers, electronic pressure transmitters or similar instrumentation.
Air Flow - Velocity and Dynamic Head Chart
The charts below are based on air density 1.205 kg/m3 and water density 1000 kg/m3.
Note that as indicated in the diagram above - pitot tubes are not suited for low velocity flow. Due to low dynamic pressure (head) the readings will be inaccurate.
Water Flow - Velocity and Dynamic Head Chart
Point Velocity Area Method for Flow Metering
The point velocities in a duct, channel or pipe can be measured by traversing the cross-sectional area of the conduit. The point velocities can be used to calculate the average velocity that can be used to estimate the flow.
The average velocity cab be calculated as
va = Σ vn / n
= (2 g Σ hn / n )1/2 (6)
va = average velocity (m/s, ft/s)
vn = point velocity (m/s, ft/s)
hn = point height difference pitot meter (m, ft)
n = number of point velocities
The volume flow in the conduit can be calculated as
q = va A (7)
q = flow (m3/s, ft3/s)
A = cross-sectional area (m2, ft2)
- en: pitot tube flow velocity speed volume measurement
- es: medición del volumen de la velocidad de la velocidad del flujo tubo de Pitot
- de: Staurohr Strömungsgeschwindigkeit Geschwindigkeit Volumenmessung