Engineering ToolBox - Resources, Tools and Basic Information for Engineering and Design of Technical Applications!

# Target - Volume Flow Meters

## Introduction to the target flow meters principles.

With the target meter a physical target is located directly in the fluid flow.

The deflection of the target and the force bar is measured in the instrument.

The force on the target can be expressed as

F =cd ρ v2 At / 2                                         (1)

where

F = force on the target (N)

cd = overall drag coefficient obtained from empirical data

ρ = density of fluid (kg/m3)

v = fluid velocity (m/s)

At = target area (m2)

### Advantages with the Target Flow Meter

• Can be used for any type of liquid, gas, or steam cryogenics.
• There is no moving parts such as bearings, to wear out causing failures.
• High reliability where life tests have been made to 20,000,000 cycles.
• Can be used for any line size from 0.5 inches and up with any type of mounting.
• Range/fluid changes accomplished by simply changing targets
• Turndowns aprox.15:1
• Can accept bi-directional flow where signal polarity indicates direction.
• Several materials are available. Among them 303/304 SS, 316 SS, hastelloy and inconel.

### Disadvantages with the Target Flow Meter

• Calibration must be verified in the field.

## Related Topics

• ### Flow Measurements

Flow metering principles - Orifice, Venturi, Flow Nozzles, Pitot Tubes, Target, Variable Area, Positive Displacement, Turbine, Vortex, Electromagnetic, Ultrasonic Doppler, Ultrasonic Time-of-travel, Mass Coriolis, Mass Thermal, Weir V-notch, Flume Parshall and Sluice Gate flow meters and more.

## Related Documents

• ### Comparing Flowmeters

A limited comparison of flowmeter principles - regarding service, rangeability, pressure loss, typical accuracy, upstream pipe diameters, viscosity and relative costs.
• ### Drag Coefficient

The drag coefficient quantifies the drag or resistance of an object in a fluid environment.
• ### Flowmeter - Accuracy

Introduction to accuracy in flow measurement devices.
• ### Flowmeters - Turndown Ratios

Turndown ratio (Rangeability) can be used to compare flow measurement devices like orifices, venturi meters etc.

## Search

Search is the most efficient way to navigate the Engineering ToolBox.

## Engineering ToolBox - SketchUp Extension - Online 3D modeling!

Add standard and customized parametric components - like flange beams, lumbers, piping, stairs and more - to your Sketchup model with the Engineering ToolBox - SketchUp Extension - enabled for use with older versions of the amazing SketchUp Make and the newer "up to date" SketchUp Pro . Add the Engineering ToolBox extension to your SketchUp Make/Pro from the Extension Warehouse !

We don't collect information from our users. More about

We use a third-party to provide monetization technologies for our site. You can review their privacy and cookie policy here.

You can change your privacy settings by clicking the following button: .

## Citation

• The Engineering ToolBox (2003). Target - Volume Flow Meters. [online] Available at: https://www.engineeringtoolbox.com/target-flow-meters-d_497.html [Accessed Day Month Year].

Modify the access date according your visit.

6.3.17

.