# Flowmeter - Accuracy

Accuracy indicates how close a measured value is to the accepted or true value.

*"Accuracy is the degree of conformance of a measurement to standard or true value"*

A measurement system consists normally of a *primary element* and a *secondary element*. Example - in a flow measurement system the orifice is the primary element and the pressure transmitter is the secondary element. Both the orifice and the pressure transmitter influence the flow measurement total accuracy.

The accuracy of flow meters can be stated as:

*percent of full span**percent of rate*

### Accuracy as Percent of Full Span

Accuracy in percent of full span can be expressed as:

a_{%}= a 100% / s(1)

where

a_{%}= relative accuracy (%)

a= +/- absolute accuracy

s= full scale of the flow meter

#### Example - Accuracy of Flow Meter in Percent of Full Span

If the absolute accuracy is *+/- 10 kg/h* - the relative accuracy at full scale *1000 kg/h* can be calculated as:

a_{%}= (+/- 10 kg/h) 100% / (1000 kg/h)

= +/- 1%

The relative accuracy at full scale *100 kg/h* can be calculated as:

a_{%}= (+/- 10 kg/h) 100% / (100 kg/h)

= +/- 10%

### Accuracy as percent of Rate

Accuracy in percent of rate can be expressed as:

a_{%}= a 100% / q(2)

where

a_{%}= relative accuracy (%)

a= +/- absolute accuracy

q= flow rate

#### Example - Accuracy of Flow Meter in Percent of Rate

If the relative accuracy is *+/- 1%* of rate - the absolute accuracy at rate *1000 kg/h *can be calculated as:

a = (+/- 1%) (1000 kg/h)

= +/- 10 kg/h

The absolute accuracy at rate *100 kg/h* can be calculated as:

a = (+/- 1%) (100 kg/h)

= +/- 1 kg/h

### Hysteresis

Hysteresis is the maximum difference between measurement readings at the same mechanical set point when the point is set from a value above the set point and reset from a value below the set point. Hysteresis can increase over time as a result of wear and tear of the mechanical parts.

### Standard Deviation Estimate

*Standard Deviation Estimate* is the measure of dispersion of a set of data in its distribution about the mean of the set and can be expressed as

*S = [Σ(q _{a} - q_{i})^{2} / (n - 1)]^{1/2} (3)*

*where*

*S = standard deviation estimate*

*q _{a} = mean value of measurements*

*q _{i} = each individual measurement value from the set*

*n = number of values in the set of measurements*

The *mean value* can be calculated as

*q _{a} = Σq_{i} / n (4)*

#### Example - Estimated Standard Deviation

The flow through a flow meter is measured to *10 kg/h, 9 kg/h and 8.5 kg/h*.

The mean flow can be calculated as

*q _{a} = [(10 kg/h) + (9 kg/h) + (8.5 kg/h)] / 3 *

* = 9.2 kg/h*

The standard deviation can be estimated as

*S = [(((9.2 kg/h) - (10 kg/h)) ^{2} + ((9.2 kg/h) - (9 kg/h))^{2} + ((9.2 kg/h) - (8.5 kg/h))^{2}) / (3 - 1)]^{1/2} *

* = 0.76 kg/h*

## Related Topics

### • Flow Measurement

Flow metering principles - Orifice, Venturi, Flow Nozzles, Pitot Tubes, Target, Variable Area, Positive Displacement, Turbine, Vortex, Electromagnetic, Ultrasonic Doppler, Ultrasonic Time-of-travel, Mass Coriolis, Mass Thermal, Weir V-notch, Flume Parshall and Sluice Gate flow meters and more.

## Related Documents

### California Pipe Flow Metering Method

Calculate the discharge length from the open end of a partially filled horizontal pipe.

### Comparing Flowmeters

A limited comparison of flowmeter principles - regarding service, rangeability, pressure loss, typical accuracy, upstream pipe diameters, viscosity and relative costs.

### Cryogenic Fluids and Liquefied Gas Properties

Cryogenic properties as density, boiling points and heat of evaporation for fluids like hydrogen, methane, oxygen, nitrogen, fluorine and helium.

### Electromagnetic Flowmeters

An introduction to electromagnetic flowmeters and their principles.

### Flowmeters - Turndown Ratios

Turndown ratio (Rangeability) can be used to compare flow measurement devices like orifices, venturi meters etc.

### Fluid Flowmeters - Comparing Types

An introduction to the different types of fluid flowmeters - Orifices, Venturies, Nozzles, Rotameters, Pitot Tubes, Calorimetrics, Turbine, Vortex, Electromagnetic, Doppler, Ultrasonic, Thermal, Coriolis.

### Orifice, Nozzle and Venturi Flow Rate Meters

The orifice, nozzle and venturi flow rate meters makes the use of the Bernoulli Equation to calculate fluid flow rate using pressure difference through obstructions in the flow.

### Pitot Tubes

Pitot tubes can be used to measure fluid flow velocities by measuring the difference between static and dynamic pressure in the flow.

### Process Instrumentation - ISA Codes

ISA process instrumentation codes - and combinations.

### Sluice Gate - Volume Flow Measurements

Sluice gates can be used to control and measure volume flow rates in open channels and rivers, mainly in connection to hydro power plants.

### Target - Volume Flow Meters

Introduction to the target flow meters principles.

### U-Tube Differential Pressure Manometers

Inclined and vertical u-tube manometers used to measure differential pressure in flow meters like pitot tubes, orifices and nozzles.

### Ultrasonic Doppler and Time of Flight Velocity and Volume Flow Meters

An basic introduction to the ultrasonic Doppler and Time of Flight Flow Meters.

### Velocity-Area Flowmetering

Flow rate or discharge in an open conduit, channel or river can be calculated with the velocity-area principle.

### Volume Flow - Online Unit Converter

Convert between volume flow units like *gpm, liter/sec, cfm, m ^{3}/h*.

### Vortex Flowmeters

An introduction to vortex flowmeters.

### Water - Volume Flow Measurement

A manual providing guidance in selecting, managing, inspecting, and maintaining water measurement devices - describes standard methods and devices commonly used to measure irrigation water.