# Flowmeter Accuracy

## Introduction to accuracy in flow measurement devices

Accuracy indicates how close a measured value is to the accepted or true value.

• "Accuracy is the degree of conformance of a measurement to standard or true value"

A measurement system consists normally of a primary element and a secondary element. Example - in a flow measurement system the orifice is the primary element and the pressure transmitter is the secondary element. Both the orifice and the pressure transmitter influence the flow measurement total accuracy.

The accuracy of flow meters can be stated as:

• percent of full span
• percent of rate

### Accuracy as Percent of Full Span

Accuracy in percent of full span can be expressed as:

a% = a 100% / s                                (1)

where

a% = relative accuracy (%)

a = +/- absolute accuracy

s = full scale of the flow meter

#### Example - Accuracy of Flow Meter in Percent of Full Span

If the absolute accuracy is +/- 10 kg/h - the relative accuracy at full scale 1000 kg/h can be calculated as:

a% = (+/- 10 kg/h) 100% / (1000 kg/h)

= +/- 1%

The relative accuracy at full scale 100 kg/h can be calculated as:

a% = (+/- 10 kg/h) 100% / (100 kg/h)

= +/- 10%

### Accuracy as percent of Rate

Accuracy in percent of rate can be expressed as:

a% = a 100% / q                                (2)

where

a% = relative accuracy (%)

a = +/- absolute accuracy

q = flow rate

#### Example - Accuracy of Flow Meter in Percent of Rate

If the relative accuracy is +/- 1% of rate - the absolute accuracy at rate 1000 kg/h can be calculated as:

a = (+/- 1%) (1000 kg/h)

= +/- 10 kg/h

The absolute accuracy at rate 100 kg/h can be calculated as:

a = (+/- 1%) (100 kg/h)

= +/- 1 kg/h

### Hysteresis

Hysteresis is the maximum difference between measurement readings at the same mechanical set point when the point is set from a value above the set point and reset from a value below the set point. Hysteresis can increase over time as a result of wear and tear of the mechanical parts.

### Standard Deviation Estimate

Standard Deviation Estimate is the measure of dispersion of a set of data in its distribution about the mean of the set and can be expressed as

S = [Σ(qa - qi)2 / (n - 1)]1/2                                      (3)

where

S = standard deviation estimate

qa = mean value of measurements

qi = each individual measurement value from the set

n = number of values in the set of measurements

The mean value can be calculated as

qa = Σqi / n                       (4)

#### Example - Estimated Standard Deviation

The flow through a flow meter is measured to 10 kg/h, 9 kg/h and 8.5 kg/h.

The mean flow can be calculated as

qa = [(10 kg/h) + (9 kg/h) + (8.5 kg/h)] / 3

= 9.2 kg/h

The standard deviation can be estimated as

S = [(((9.2 kg/h) - (10 kg/h))2 + ((9.2 kg/h) - (9 kg/h))2 + ((9.2 kg/h) - (8.5 kg/h))2) / (3 - 1)]1/2

= 0.76 kg/h

## Related Topics

• Flow Measurement - Flow metering principles - Orifice, Venturi, Flow Nozzles, Pitot Tubes, Target, Variable Area, Positive Displacement, Turbine, Vortex, Electromagnetic, Ultrasonic Doppler, Ultrasonic Time-of-travel, Mass Coriolis, Mass Thermal, Weir V-notch, Flume Parshall and Sluice Gate flow meters and more

## Tag Search

• en: flowmeter accuracy full span measurement
• es: exactitud de mediciĆ³n de caudal envergadura total
• de: Durchflussmesser Genauigkeit vollSpanne Mess

## Search the Engineering ToolBox

- "the most efficient way to navigate!"

## Engineering ToolBox - SketchUp Extension - Online 3D modeling!

Add standard and customized parametric components - like flange beams, lumbers, piping, stairs and more - to your SketchUp model with the Engineering ToolBox - SketchUp Extension/Plugin - enabled for use with the amazing, fun and free SketchUp Make and SketchUp Pro . Add the Engineering ToolBox extension to your SketchUp from the Sketchup Extension Warehouse!

Translate the Engineering ToolBox!