Engineering ToolBox - Resources, Tools and Basic Information for Engineering and Design of Technical Applications!

# Instruments - Static Characteristics

## Characteristics that change very slowly with time

The static characteristics of instruments are attributes that changes slowly with time. Static characteristics can be divided in to desirable and undesirable.

Desirable characteristics - what we want to achieve - are

• Accuracy
• Sensitivity
• Repeatability
• Reproducibility

Undesirable characteristics - what we want to avoid - are

• Drift
• Threshold
• Hystersis
• Creep
• Resolution
• Static error

### Desirable Characteristics

#### Accuracy

Accuracy is

• the closeness of a measurement to the true value

Relative accuracy can be expressed as

ar = (ymax - x) / x                                       (1)

where

ar = relative accuracy (unit/unit)

x = input true value (unit)

y = instrument output (unit)

##### Example - Accuracy

The true length of a steel beam is 6 m. Three repeated readings with a laser meter indicates a length of 6.01 m, 6.0095 and 6.015 m. The accuracy based on maximum difference can be calculated as

an absolute value like

aa = 6.015 m - 6 m

= 0.015 m

or as a relative value

ar = (6.015 m - 6 m) / 6 m

= 0.0025 m/m

or as relative value in percentage

a% = ((6.015 m - 6 m) / 6 m) 100%

= 0.25 %

The accuracy of an instrument can be related to

• maximum measured value possible for the instrument
• maximum range for the instrument
• actual output from the instrument

Two terms commonly used in connection with accuracy are precision, trueness and calibration.

##### Precision
• Precision is the closeness of agreement among a set of results

Example - for the steel beam above all measurements are within ±0.01 m - and we could say that the precision is good.

##### Trueness
• Trueness is the closeness of the mean of a set of measurement results to the actual (true) value

Example - for the steel beam above the mean value of the set of measurements is 6.01 m - and we could say that the trueness could have been better.

##### Calibration

The precision of the laser meter used in the example above is good and the accuracy of the meter can be improved with calibration as

calibration = 6.01 m - 6 m

= 0.01 m

#### Sensitivity

Sensitivity is the increment of the output signal (or response) to the increment of the input measured signal - and can be expressed as

s = dy / dx                           (2)

where

s = sensitivity  (output unit / input unit)

dy = change instrument output value (output unit)

dx = change in input true value (input unit)

##### Example - Temperature measurement with a Pt100 Platinum Resistance Thermometer

When temperature is changed from 0oC to 50oC - the resistance in a Pt100 thermometer changes from 100 ohm to 119.4 ohm. The sensitivity for this range can be calculated as

s = (119.4 ohm - 100 ohm) / (50oC - 0oC)

=  0.388 ohm/oC

#### Repeatability

Repeatability is the variation in measurements taken on the same item under the same conditions.

#### Reproducibility

Reproducibility is the ability of a measurement to be duplicated, either by the same person or by someone else under changed conditions.

### Undesirable Characteristics

#### Drift

Drift is the change in instrument output over time - when the true value is constant.

Dead zone errors are created by

##### Threshold

Threshold is when a minimum input is required to generate change in output.

##### Hystersis

An example can be a nut that is screwed a number of turns on a threaded rod. When turned back the same number of turns the nut will not be in the exact the same position as at the start. This is a typical a problem that must be adressed in applications like cnc machines and 3d printers.

##### Creep

Creep is caused by the time an instrument need to adapt to change in aplied input.

##### Resolution

Depending on the instrument - but minimum change in input may required for change in output.

### Significant Figures

Uncertainty in measurements or calculated values is indicated by the number of significant figures.

• Significant figures are the digits of a number known with certain certainty

The last digit in a number is taken as uncertain while the other digits are regarded as certain.

#### Example - Significant Figures

For the number 31.2 we regard the two first digits 30 as certain and the last digit 6 as uncertain. Unless otherwise stated an uncertainity of ±1 is assumed for the last digit.

## Related Topics

• ### Measurements and Instrumentation

Measurement and instrumentation strategies.

## Search

Search is the most efficient way to navigate the Engineering ToolBox.

## Engineering ToolBox - SketchUp Extension - Online 3D modeling!

Add standard and customized parametric components - like flange beams, lumbers, piping, stairs and more - to your Sketchup model with the Engineering ToolBox - SketchUp Extension - enabled for use with older versions of the amazing SketchUp Make and the newer "up to date" SketchUp Pro . Add the Engineering ToolBox extension to your SketchUp Make/Pro from the Extension Warehouse !

We don't collect information from our users. More about

We use a third-party to provide monetization technologies for our site. You can review their privacy and cookie policy here.

You can change your privacy settings by clicking the following button: .

## Citation

• The Engineering ToolBox (2017). Instruments - Static Characteristics. [online] Available at: https://www.engineeringtoolbox.com/instruments-static-characteristics-d_1985.html [Accessed Day Month Year].

Modify the access date according your visit.

6.3.17

.